Discomfort with multifocal soft contact lenses due to moderate dry eye syndrome.
|This 56-year-old female patient presented for a scleral lens fitting with a history of
refractive error and moderate dry eye syndrome.
She is a soft multifocal contact lens wearer who was not achieving her desired level of
acuity. In addition, her moderate dry eye syndrome impacts her ability to wear contact
lenses and greatly impacts her visual quality and stability.
||Patient’s Uncorrected VA (distance and near) is:
DVA: OD: 20/400, OS: 20/400, OU: 20/200
NVA: OD: 20/20, OS: 20/20, OU: 20/20
Patient’s ocular shape was evaluated using the Pentacam CSP saggital height map.
shows CSP reports for both eyes. There is an apparent moderate scleral toricity OU, indicating
that the patient is likely to benefit from a toric or a freeform scleral lens design.
The initial lens power was determined by over-refracting the patient with a 7.85-mm BC
Valley Contax Custom Stable Elite scleral lens.
LENS SELECTION AND DESIGN
The patient was fitted with a GaudiTM
scleral lens. Since the patient’s HVID was 12.5-mm OU we
opted for a 16.8-mm diameter lenses with a moderate fit factor to gently match the contours of the lens to the
shape of the eye.
shows the predicted clearance maps calculated using the Gaudi lens design
software and the radial saggital profiles (red line) compared to the CSP measurement (black line) at the
First lens set based on CSP data. No remakes.
DVA: OD: 20/20-1, OS: 20/20, OU: 20/20
NVA: OU: 20/25+
Excellent lens comfort and vision throughout 8+ hours of wear. Patient reports vision, distance and near, is
better quality and stability
than her current daily disposable multifocal lenses.
Central clearance was measured using an AS OCT. The reported values were 356-µm OD and 327-µm OS
), which after correcting for the
refractive index of the tear film give: 270-µm OD and 245-µm OS.
shows the Gaudi lens landing zone
and edge profile. Figure 5
shows great lens alignment
and no impingement.
The patient achieved improved comfort with lens wear and has been able to wear her lenses more throughout the
day compared to her
daily disposable lenses. She is much happier with her vision and there has been immense improvement to her
visual quality and stability.
The fitting process was significantly simplified with the use of the Pentacam CSP module and Gaudi Scleral
lens design software. Excellent comfort and VA were achieved with the first lens without remakes.